ncnrdata  NCNR Data¶
ANDR 
Instrument definition for NCNR AND/R diffractometer/reflectometer. 
MAGIK 
Instrument definition for NCNR MAGIK diffractometer/reflectometer. 
NCNRData 

NG1 
Instrument definition for NCNR NG1 reflectometer. 
NG7 
Instrument definition for NCNR NG7 reflectometer. 
PBR 
Instrument definition for NCNR PBR reflectometer. 
XRay 
Instrument definition for NCNR Xray reflectometer. 
find_xsec 
Find files containing the polarization crosssections. 
load 
Return a probe for NCNR data. 
load_magnetic 
Return a probe for magnetic NCNR data. 
parse_ncnr_file 
Parse NCNR reduced data file returning header and data. 
NCNR data loaders
The following instruments are defined:
MAGIK, PBR, ANDR, NG1, NG7 and XRay
These are refl1d.instrument.Monochromatic
classes tuned with default
instrument parameters and loaders for reduced NCNR data.
The instruments can be used to load data or to compute resolution functions for the purposes.
Example loading data:
>>> from refl1d.names import *
>>> datafile = sample_data('chale207.refl')
>>> instrument = NCNR.ANDR(Tlo=0.5, slits_at_Tlo=0.2, slits_below=0.1)
>>> probe = instrument.load(datafile)
>>> probe.plot(view='log')
Magnetic data has multiple cross sections and often has fixed slits:
>>> datafile = sample_data('lha03_255G.refl')
>>> instrument = NCNR.NG1(slits_at_Tlo=1)
>>> probe = instrument.load_magnetic(datafile)
>>> probe.plot(view='SA', substrate=silicon) # Spin asymmetry view
For simulation, you need a probe and a sample:
>>> instrument = NCNR.ANDR(Tlo=0.5, slits_at_Tlo=0.2, slits_below=0.1)
>>> probe = instrument.probe(T=numpy.linspace(0,5,51))
>>> probe.plot_resolution()
>>> sample = silicon(0,10)  gold(100,10)  air
>>> M = Experiment(probe=probe, sample=sample)
>>> M.simulate_data() # Optional
>>> M.plot()
And for magnetic:
>>> instrument = NCNR.NG1(slits_at_Tlo=1)
>>> #sample = silicon(0,10)  Magnetic(permalloy(100,10),rho_M=3)  air
>>> #M = Experiment(probe=probe, sample=sample)
>>> #M.simulate_data()
>>> #M.plot()
>>> #probe = instrument.simulate_magnetic(sample, T=numpy.linspace(0,5,51))
>>> #h = pylab.plot(probe.Q, probe.dQ)
>>> #h = pylab.ylabel('resolution (1sigma)')
>>> #h = pylab.xlabel('Q (inv A)')
See instrument
for details.

class
refl1d.ncnrdata.
ANDR
(**kw)[source]¶ Bases:
refl1d.ncnrdata.NCNRData
,refl1d.instrument.Monochromatic
Instrument definition for NCNR AND/R diffractometer/reflectometer.

Thi
= 90¶

Tlo
= 90¶

calc_dT
(**kw)¶ Compute the angular divergence for given slits and angles
Parameters:  T OR Q : [float]  ° OR Å^{1}
measurement angles
 slits : float OR (float,float)  mm
total slit opening from edge to edge, not beam center to edge
 d_s1, d_s2 : float  mm
distance from sample to slit 1 and slit 2
 sample_width : float  mm
size of sample
 sample_broadening : float  ° FWHM
resolution changes from sample warp
Returns:  dT : [float]  ° FWHM
angular divergence
sample_broadening can be estimated from W, the full width at half maximum of a rocking curve measured in degrees:
sample_broadening = W  degrees( 0.5*(s1+s2) / (d1d2))

calc_slits
(**kw)¶ Determines slit openings from measurement pattern.
If slits are fixed simply return the same slits for every angle, otherwise use an opening range [Tlo,Thi] and the value of the slits at the start of the opening to define the slits. Slits below Tlo and above Thi can be specified separately.
T OR Q incident angle or Q Tlo, Thi angle range over which slits are opening slits_at_Tlo openings at the start of the range, or fixed opening slits_below, slits_above openings below and above the range
Use fixed_slits is available, otherwise use opening slits.

dLoL
= 0.009¶

d_s1
= 2086.0¶

d_s2
= 230.0¶

defaults
()¶ Return default instrument properties as a printable string.

fixed_slits
= None¶

instrument
= 'AND/R'¶

load
(filename, **kw)¶

load_magnetic
(filename, **kw)¶

magnetic_probe
(Aguide=270, shared_beam=True, H=0, **kw)¶ Simulate a polarized measurement probe.
Returns a probe with Q, angle, wavelength and the associated uncertainties, but not any data.
Guide field angle Aguide can be specified, as well as keyword arguments for the geometry of the probe cross sections such as slits_at_Tlo, Tlo, Thi, slits_below, and slits_above to define the angular divergence.

probe
(**kw)¶ Return a probe for use in simulation.
Parameters:  Q : [float]  Å
Q values to be measured.
 T : [float]  °
Angles to be measured.
Additional keyword parameters
Returns:  probe : Probe
Measurement probe with complete resolution information. The probe will not have any data.
If both Q and T are specified then Q takes precedents.
You can override instrument parameters using key=value. In particular, settings for slits_at_Tlo, Tlo, Thi, slits_below, and slits_above are used to define the angular divergence.

radiation
= 'neutron'¶

readfile
(filename)¶

resolution
(**kw)¶ Calculate resolution at each angle.
Return:  T, dT : [float]  °
Angles and angular divergence.
 L, dL : [float]  Å
Wavelengths and wavelength dispersion.

sample_broadening
= 0¶

sample_width
= 10000000000.0¶

slits_above
= None¶

slits_at_Tlo
= None¶

slits_below
= None¶

wavelength
= 5.0042¶


class
refl1d.ncnrdata.
MAGIK
(**kw)[source]¶ Bases:
refl1d.ncnrdata.NCNRData
,refl1d.instrument.Monochromatic
Instrument definition for NCNR MAGIK diffractometer/reflectometer.

Thi
= 90¶

Tlo
= 90¶

calc_dT
(**kw)¶ Compute the angular divergence for given slits and angles
Parameters:  T OR Q : [float]  ° OR Å^{1}
measurement angles
 slits : float OR (float,float)  mm
total slit opening from edge to edge, not beam center to edge
 d_s1, d_s2 : float  mm
distance from sample to slit 1 and slit 2
 sample_width : float  mm
size of sample
 sample_broadening : float  ° FWHM
resolution changes from sample warp
Returns:  dT : [float]  ° FWHM
angular divergence
sample_broadening can be estimated from W, the full width at half maximum of a rocking curve measured in degrees:
sample_broadening = W  degrees( 0.5*(s1+s2) / (d1d2))

calc_slits
(**kw)¶ Determines slit openings from measurement pattern.
If slits are fixed simply return the same slits for every angle, otherwise use an opening range [Tlo,Thi] and the value of the slits at the start of the opening to define the slits. Slits below Tlo and above Thi can be specified separately.
T OR Q incident angle or Q Tlo, Thi angle range over which slits are opening slits_at_Tlo openings at the start of the range, or fixed opening slits_below, slits_above openings below and above the range
Use fixed_slits is available, otherwise use opening slits.

dLoL
= 0.009¶

d_s1
= 1759.0¶

d_s2
= 330.0¶

defaults
()¶ Return default instrument properties as a printable string.

fixed_slits
= None¶

instrument
= 'MAGIK'¶

load
(filename, **kw)¶

load_magnetic
(filename, **kw)¶

magnetic_probe
(Aguide=270, shared_beam=True, H=0, **kw)¶ Simulate a polarized measurement probe.
Returns a probe with Q, angle, wavelength and the associated uncertainties, but not any data.
Guide field angle Aguide can be specified, as well as keyword arguments for the geometry of the probe cross sections such as slits_at_Tlo, Tlo, Thi, slits_below, and slits_above to define the angular divergence.

probe
(**kw)¶ Return a probe for use in simulation.
Parameters:  Q : [float]  Å
Q values to be measured.
 T : [float]  °
Angles to be measured.
Additional keyword parameters
Returns:  probe : Probe
Measurement probe with complete resolution information. The probe will not have any data.
If both Q and T are specified then Q takes precedents.
You can override instrument parameters using key=value. In particular, settings for slits_at_Tlo, Tlo, Thi, slits_below, and slits_above are used to define the angular divergence.

radiation
= 'neutron'¶

readfile
(filename)¶

resolution
(**kw)¶ Calculate resolution at each angle.
Return:  T, dT : [float]  °
Angles and angular divergence.
 L, dL : [float]  Å
Wavelengths and wavelength dispersion.

sample_broadening
= 0¶

sample_width
= 10000000000.0¶

slits_above
= None¶

slits_at_Tlo
= None¶

slits_below
= None¶

wavelength
= 5.0042¶


class
refl1d.ncnrdata.
NG1
(**kw)[source]¶ Bases:
refl1d.ncnrdata.NCNRData
,refl1d.instrument.Monochromatic
Instrument definition for NCNR NG1 reflectometer.

Thi
= 90¶

Tlo
= 90¶

calc_dT
(**kw)¶ Compute the angular divergence for given slits and angles
Parameters:  T OR Q : [float]  ° OR Å^{1}
measurement angles
 slits : float OR (float,float)  mm
total slit opening from edge to edge, not beam center to edge
 d_s1, d_s2 : float  mm
distance from sample to slit 1 and slit 2
 sample_width : float  mm
size of sample
 sample_broadening : float  ° FWHM
resolution changes from sample warp
Returns:  dT : [float]  ° FWHM
angular divergence
sample_broadening can be estimated from W, the full width at half maximum of a rocking curve measured in degrees:
sample_broadening = W  degrees( 0.5*(s1+s2) / (d1d2))

calc_slits
(**kw)¶ Determines slit openings from measurement pattern.
If slits are fixed simply return the same slits for every angle, otherwise use an opening range [Tlo,Thi] and the value of the slits at the start of the opening to define the slits. Slits below Tlo and above Thi can be specified separately.
T OR Q incident angle or Q Tlo, Thi angle range over which slits are opening slits_at_Tlo openings at the start of the range, or fixed opening slits_below, slits_above openings below and above the range
Use fixed_slits is available, otherwise use opening slits.

dLoL
= 0.015¶

d_s1
= 1905.0¶

d_s2
= 355.59999999999997¶

d_s3
= 228.6¶

d_s4
= 1066.8¶

defaults
()¶ Return default instrument properties as a printable string.

fixed_slits
= None¶

instrument
= 'NG1'¶

load
(filename, **kw)¶

load_magnetic
(filename, **kw)¶

magnetic_probe
(Aguide=270, shared_beam=True, H=0, **kw)¶ Simulate a polarized measurement probe.
Returns a probe with Q, angle, wavelength and the associated uncertainties, but not any data.
Guide field angle Aguide can be specified, as well as keyword arguments for the geometry of the probe cross sections such as slits_at_Tlo, Tlo, Thi, slits_below, and slits_above to define the angular divergence.

probe
(**kw)¶ Return a probe for use in simulation.
Parameters:  Q : [float]  Å
Q values to be measured.
 T : [float]  °
Angles to be measured.
Additional keyword parameters
Returns:  probe : Probe
Measurement probe with complete resolution information. The probe will not have any data.
If both Q and T are specified then Q takes precedents.
You can override instrument parameters using key=value. In particular, settings for slits_at_Tlo, Tlo, Thi, slits_below, and slits_above are used to define the angular divergence.

radiation
= 'neutron'¶

readfile
(filename)¶

resolution
(**kw)¶ Calculate resolution at each angle.
Return:  T, dT : [float]  °
Angles and angular divergence.
 L, dL : [float]  Å
Wavelengths and wavelength dispersion.

sample_broadening
= 0¶

sample_width
= 10000000000.0¶

slits_above
= None¶

slits_at_Tlo
= None¶

slits_below
= None¶

wavelength
= 4.75¶


class
refl1d.ncnrdata.
NG7
(**kw)[source]¶ Bases:
refl1d.ncnrdata.NCNRData
,refl1d.instrument.Monochromatic
Instrument definition for NCNR NG7 reflectometer.

Thi
= 90¶

Tlo
= 90¶

calc_dT
(**kw)¶ Compute the angular divergence for given slits and angles
Parameters:  T OR Q : [float]  ° OR Å^{1}
measurement angles
 slits : float OR (float,float)  mm
total slit opening from edge to edge, not beam center to edge
 d_s1, d_s2 : float  mm
distance from sample to slit 1 and slit 2
 sample_width : float  mm
size of sample
 sample_broadening : float  ° FWHM
resolution changes from sample warp
Returns:  dT : [float]  ° FWHM
angular divergence
sample_broadening can be estimated from W, the full width at half maximum of a rocking curve measured in degrees:
sample_broadening = W  degrees( 0.5*(s1+s2) / (d1d2))

calc_slits
(**kw)¶ Determines slit openings from measurement pattern.
If slits are fixed simply return the same slits for every angle, otherwise use an opening range [Tlo,Thi] and the value of the slits at the start of the opening to define the slits. Slits below Tlo and above Thi can be specified separately.
T OR Q incident angle or Q Tlo, Thi angle range over which slits are opening slits_at_Tlo openings at the start of the range, or fixed opening slits_below, slits_above openings below and above the range
Use fixed_slits is available, otherwise use opening slits.

dLoL
= 0.025¶

d_detector
= 2000.0¶

d_s1
= 1625.0¶

d_s2
= 275.0¶

defaults
()¶ Return default instrument properties as a printable string.

fixed_slits
= None¶

instrument
= 'NG7'¶

load
(filename, **kw)¶

load_magnetic
(filename, **kw)¶

magnetic_probe
(Aguide=270, shared_beam=True, H=0, **kw)¶ Simulate a polarized measurement probe.
Returns a probe with Q, angle, wavelength and the associated uncertainties, but not any data.
Guide field angle Aguide can be specified, as well as keyword arguments for the geometry of the probe cross sections such as slits_at_Tlo, Tlo, Thi, slits_below, and slits_above to define the angular divergence.

probe
(**kw)¶ Return a probe for use in simulation.
Parameters:  Q : [float]  Å
Q values to be measured.
 T : [float]  °
Angles to be measured.
Additional keyword parameters
Returns:  probe : Probe
Measurement probe with complete resolution information. The probe will not have any data.
If both Q and T are specified then Q takes precedents.
You can override instrument parameters using key=value. In particular, settings for slits_at_Tlo, Tlo, Thi, slits_below, and slits_above are used to define the angular divergence.

radiation
= 'neutron'¶

readfile
(filename)¶

resolution
(**kw)¶ Calculate resolution at each angle.
Return:  T, dT : [float]  °
Angles and angular divergence.
 L, dL : [float]  Å
Wavelengths and wavelength dispersion.

sample_broadening
= 0¶

sample_width
= 10000000000.0¶

slits_above
= None¶

slits_at_Tlo
= None¶

slits_below
= None¶

wavelength
= 4.768¶


class
refl1d.ncnrdata.
PBR
(**kw)[source]¶ Bases:
refl1d.ncnrdata.NCNRData
,refl1d.instrument.Monochromatic
Instrument definition for NCNR PBR reflectometer.

Thi
= 90¶

Tlo
= 90¶

calc_dT
(**kw)¶ Compute the angular divergence for given slits and angles
Parameters:  T OR Q : [float]  ° OR Å^{1}
measurement angles
 slits : float OR (float,float)  mm
total slit opening from edge to edge, not beam center to edge
 d_s1, d_s2 : float  mm
distance from sample to slit 1 and slit 2
 sample_width : float  mm
size of sample
 sample_broadening : float  ° FWHM
resolution changes from sample warp
Returns:  dT : [float]  ° FWHM
angular divergence
sample_broadening can be estimated from W, the full width at half maximum of a rocking curve measured in degrees:
sample_broadening = W  degrees( 0.5*(s1+s2) / (d1d2))

calc_slits
(**kw)¶ Determines slit openings from measurement pattern.
If slits are fixed simply return the same slits for every angle, otherwise use an opening range [Tlo,Thi] and the value of the slits at the start of the opening to define the slits. Slits below Tlo and above Thi can be specified separately.
T OR Q incident angle or Q Tlo, Thi angle range over which slits are opening slits_at_Tlo openings at the start of the range, or fixed opening slits_below, slits_above openings below and above the range
Use fixed_slits is available, otherwise use opening slits.

dLoL
= 0.015¶

d_s1
= 1835¶

d_s2
= 343¶

d_s3
= 380¶

d_s4
= 1015¶

defaults
()¶ Return default instrument properties as a printable string.

fixed_slits
= None¶

instrument
= 'PBR'¶

load
(filename, **kw)¶

load_magnetic
(filename, **kw)¶

magnetic_probe
(Aguide=270, shared_beam=True, H=0, **kw)¶ Simulate a polarized measurement probe.
Returns a probe with Q, angle, wavelength and the associated uncertainties, but not any data.
Guide field angle Aguide can be specified, as well as keyword arguments for the geometry of the probe cross sections such as slits_at_Tlo, Tlo, Thi, slits_below, and slits_above to define the angular divergence.

probe
(**kw)¶ Return a probe for use in simulation.
Parameters:  Q : [float]  Å
Q values to be measured.
 T : [float]  °
Angles to be measured.
Additional keyword parameters
Returns:  probe : Probe
Measurement probe with complete resolution information. The probe will not have any data.
If both Q and T are specified then Q takes precedents.
You can override instrument parameters using key=value. In particular, settings for slits_at_Tlo, Tlo, Thi, slits_below, and slits_above are used to define the angular divergence.

radiation
= 'neutron'¶

readfile
(filename)¶

resolution
(**kw)¶ Calculate resolution at each angle.
Return:  T, dT : [float]  °
Angles and angular divergence.
 L, dL : [float]  Å
Wavelengths and wavelength dispersion.

sample_broadening
= 0¶

sample_width
= 10000000000.0¶

slits_above
= None¶

slits_at_Tlo
= None¶

slits_below
= None¶

wavelength
= 4.75¶


class
refl1d.ncnrdata.
XRay
(**kw)[source]¶ Bases:
refl1d.ncnrdata.NCNRData
,refl1d.instrument.Monochromatic
Instrument definition for NCNR Xray reflectometer.
Normal dT is in the range 2e5 to 3e4.
Slits are fixed throughout the experiment in one of a few preconfigured openings. Please update this file with the standard configurations when you find them.
You can choose to ignore the geometric calculation entirely by setting the slit opening to 0 and using sample_broadening to define the entire divergence:
>>> from refl1d.names import * >>> file = sample_data("spin_valve01.refl") >>> xray = NCNR.XRay(slits_at_Tlo=0) >>> data = xray.load(file, sample_broadening=1e4) >>> print(data.dT[5]) 0.0001

Thi
= 90¶

Tlo
= 90¶

calc_dT
(**kw)¶ Compute the angular divergence for given slits and angles
Parameters:  T OR Q : [float]  ° OR Å^{1}
measurement angles
 slits : float OR (float,float)  mm
total slit opening from edge to edge, not beam center to edge
 d_s1, d_s2 : float  mm
distance from sample to slit 1 and slit 2
 sample_width : float  mm
size of sample
 sample_broadening : float  ° FWHM
resolution changes from sample warp
Returns:  dT : [float]  ° FWHM
angular divergence
sample_broadening can be estimated from W, the full width at half maximum of a rocking curve measured in degrees:
sample_broadening = W  degrees( 0.5*(s1+s2) / (d1d2))

calc_slits
(**kw)¶ Determines slit openings from measurement pattern.
If slits are fixed simply return the same slits for every angle, otherwise use an opening range [Tlo,Thi] and the value of the slits at the start of the opening to define the slits. Slits below Tlo and above Thi can be specified separately.
T OR Q incident angle or Q Tlo, Thi angle range over which slits are opening slits_at_Tlo openings at the start of the range, or fixed opening slits_below, slits_above openings below and above the range
Use fixed_slits is available, otherwise use opening slits.

dLoL
= 0.0006486766995329528¶

d_detector
= None¶

d_s1
= 275.5¶

d_s2
= 192.5¶

d_s3
= 175.0¶

defaults
()¶ Return default instrument properties as a printable string.

fixed_slits
= None¶

instrument
= 'Xray'¶

load
(filename, **kw)¶

load_magnetic
(filename, **kw)¶

magnetic_probe
(Aguide=270, shared_beam=True, H=0, **kw)¶ Simulate a polarized measurement probe.
Returns a probe with Q, angle, wavelength and the associated uncertainties, but not any data.
Guide field angle Aguide can be specified, as well as keyword arguments for the geometry of the probe cross sections such as slits_at_Tlo, Tlo, Thi, slits_below, and slits_above to define the angular divergence.

probe
(**kw)¶ Return a probe for use in simulation.
Parameters:  Q : [float]  Å
Q values to be measured.
 T : [float]  °
Angles to be measured.
Additional keyword parameters
Returns:  probe : Probe
Measurement probe with complete resolution information. The probe will not have any data.
If both Q and T are specified then Q takes precedents.
You can override instrument parameters using key=value. In particular, settings for slits_at_Tlo, Tlo, Thi, slits_below, and slits_above are used to define the angular divergence.

radiation
= 'xray'¶

readfile
(filename)¶

resolution
(**kw)¶ Calculate resolution at each angle.
Return:  T, dT : [float]  °
Angles and angular divergence.
 L, dL : [float]  Å
Wavelengths and wavelength dispersion.

sample_broadening
= 0¶

sample_width
= 10000000000.0¶

slits_above
= None¶

slits_at_Tlo
= None¶

slits_below
= None¶

wavelength
= 1.5416¶


refl1d.ncnrdata.
find_xsec
(filename)[source]¶ Find files containing the polarization crosssections.
Returns tuple with file names for ++ + + – cross sections, or None if the spin cross section does not exist.

refl1d.ncnrdata.
load
(filename, instrument=None, **kw)[source]¶ Return a probe for NCNR data.
Keyword arguments are as specified Monochromatic instruments.

refl1d.ncnrdata.
load_magnetic
(filename, Aguide=270, H=0, shared_beam=True, **kw)[source]¶ Return a probe for magnetic NCNR data.
 filename (string, or 4x string)
 If it is a string, then filenameA, filenameB, filenameC, filenameD, are the –, +, +, ++ cross sections, otherwise the individual cross sections should the be the file name for the cross section or None if the cross section does not exist.
 Aguide (degrees)
 Angle of the guide field relative to the beam. 270 is the default.
 shared_beam (True)
 Use false if beam parameters should be fit separately for the individual cross sections.
Other keyword arguments are for the individual cross section loaders as specified in
instrument.Monochromatic
.The data sets should are the base filename with an additional character corresponding to the spin state:
'a' corresponds to spin  'b' corresponds to spin + 'c' corresponds to spin + 'd' corresponds to spin ++
Unfortunately the interpretation is a little more complicated than this as the data acquisition system assigns letter on the basis of flipper state rather than neutron spin state. Whether flipper on or off corresponds to spin up or down depends on whether the polarizer/analyzer is a supermirror in transmission or reflection mode, or in the case of ^3He polarizers, whether the polarization is up or down.
For full control, specify filename as a list of files, with None for the missing cross sections.

refl1d.ncnrdata.
parse_ncnr_file
(filename)[source]¶ Parse NCNR reduced data file returning header and data.
header dictionary of fields such as ‘data’, ‘title’, ‘instrument’ data 2D array of data
If ‘columns’ is present in header, it will be a list of the names of the columns. If ‘instrument’ is present in the header, the default instrument geometry will be specified.
Slit geometry is set to the default from the instrument if it is not available in the reduced file.